1.1 This test method describes an accurate technique for measuring the normal spectral emittance of electrically nonconducting materials in the temperature range from 1000 to 1800 K, and at wavelengths from 1 to 35 ?m. It is particularly suitable for measuring the normal spectral emittance of materials such as ceramic oxides, which have relatively low thermal conductivity and are translucent to appreciable depths (several millimetres) below the surface, but which become essentially opaque at thicknesses of 10 mm or less.
1.2 This test method requires expensive equipment and rather elaborate precautions, but produces data that are accurate to within a few percent. It is particularly suitable for research laboratories, where the highest precision and accuracy are desired, and is not recommended for routine production or acceptance testing. Because of its high accuracy, this test method may be used as a reference method to be applied to production and acceptance testing in case of dispute.
1.3 This test method requires the use of a specific specimen size and configuration, and a specific heating and viewing technique. The design details of the critical specimen furnace are presented in Ref (1), and the use of a furnace of this design is necessary to comply with this test method. The transfer optics and spectrophotometer are discussed in general terms.
1.4 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
1.5 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
2. Referenced Documents
Full document current and on sale at the ASTM web store.
You are browsing archives for
Tag: THERMAL RADIATION
From the Energy Citations Database, OSTI IdentifierOSTI ID: 4830164
Technical Report, WADC-TR-59-510(Pt.III), National Bureau of Standards, Washington, D.C.,1961 Sep 01
The equipment for direct measurement of normal spectral emittance was extensively modified by incorporation of a new external optical system that increased the amount of radiant energy available for measurement by a factor of about 10, and other associated changes. The test procedure was modified by incorporation of a zero line” correction. The equipment was calibrated by means of sector-disk attenuators which passed known fractions of the radiant flux from a blackbody furnace. Working standards of normal spectral emittance were prepared, calibrated, and shipped. An equation relating the normal spectral emissivity of a metal to five other parameters of the metal, each of which makes a non-linear contribution to the emissivity, was solved for one set of data by long hand” methods. Some progress was made in setting up a program for solution of the equation by use of an electronic computer. Equipment for the automatic recording of spectral emittance data in a form suitable for direct entry into an electronic computer, and on-line computation from spectral emittance data of total emittance or solar absorptance, was designed. Specifications for the equipment were prepared and bids received preparatory to placing an order for its procurement. (auth)